Availability: In stock
|Product sold in cases of 25 tests|
|$3.92 each||$3.84 each||$3.80 each||$3.72 each|
|* Discounts may vary based on options above|
|PSC-DOA-6125M||AMP / BAR / BUP / BZO / COC / mAMP / MDMA / MTD / OPI / OXY / PCP / THC|
|Application Type||urine cup|
|Avg Competitor Price||$137.20|
PERFORMING THE TEST
Read the instructions completely.
Check the expiration date on the box. Do not use the test if it is expired. Have a watch, clock or timer ready.
Step 1: Tear the foil bag open, remove the test cup and disposable gloves provided for donor. Label the device with donor information. (Fig. 1)
Step 2: Open test cup lid. Urinate directly into the test cup. Be sure to fill the test cup with the urine specimen provided between a minimum of 30ml to maximum 110ml (marked on the cup). (Fig. 2)
Step 3: After urine specimen has been collected, close the lid securely and return cup to collection official. (Fig. 3)
Step 4: Collection official should use the gloves provided. Peel off the label to reveal test result. Read the test result at 5 minutes. DO NOT INTERPRET RESULT AFTER 10 MINUTES. (Fig. 4&5)
INTERPRETATION OF RESULTS
NEGATIVE: Two colored bands appear on the membrane. One band appears in the control region (C) and another band appears in the test region (T). Certain lines may appear lighter or thinner than other lines. A line is to be considered a line whether it is faint, light or dark.
*NOTE: The shade of red in the test line region (Drug/T) will vary, but it should be considered negative whenever there is even a faint pink line.
Preliminary Positive Result:
POSITIVE: Only one colored band appears, in the control region (C). No apparent colored band appears in the test region (T).
INVALID: Control band fails to appear. Results from any test which has not produced a control band at the specific read time must be discarded. Please review the procedure and repeat with a new test. If the problem persists, discontinue using the kit immediately and contact your local distributor.
STORAGE AND STABILITY
Store as packaged in the sealed pouch at room temperature. The test device is stable through the expiration date printed on the sealed pouch.
The test device must remain in the sealed pouch until use. DO NOT FREEZE. Do not use beyond the expiration date.
|PSC-DOA-6125M||AMP1000, BAR300, BUP10, BZO300, COC300, MET1000, MDMA500, MTD300, MOP300, OXY100, PCP25, THC50|
|Drug Name||ABB||Cutoff Level||Approx Detection|
time in Urine
time in Saliva
|Amphetamine||AMP||500 ng/ml||2-4 Days||1-3 Days|
|1000 ng/ml||2-4 Days||N/A|
|Barbiturates||BAR||300 ng/ml||3-6 Days||1 Day|
|Benzodiazepines||BZO||300 ng/ml||3-7 Days||1-2 Days|
|Buprenorphine||BUP||10 ng/ml||2-3 Days||1 Day|
|Cocaine||COC||150 ng/ml||2-4 Days||N/A|
|300 ng/ml||2-4 Days||1-3 Days|
|Ecstasy||MDMA||500 ng/ml||1-3 Days||1-2 Days|
|Fentanyl||FEN||300 ng/ml||3 Days||N/A|
|K2 / Spice||K2/Spice||50 ng/ml||1-3 Days||N/A|
|Marijuana||THC||50 ng/ml||5-30 Days||1 Day|
|20 ng/ML||Up To 30 days||N/A|
|Methadone||MTD||50 ng/ml||3-5 Days||1 Day|
|Methamphetamine||MAMP||1000 ng/ml||3-5 Days||1-3 Days|
|500 ng/ml||3-5 Days||N/A|
|Morphine||MOP||300 ng/ml||2-4 Days||1-3 Days|
|Opiates||OPI||2000 ng/ml||2-4 Days||1-3 Days|
|Oxycodone||OXY||100 ng/ml||2-4 Days||1 Day|
|Phencyclidine||PCP||25 ng/ml||5-10 Days||1-3 Days|
|Tricyclic Antidepressants||TCA||1000 ng/ml||3-6 Days||N/A|
|Propoxyphene||PPX||300 ng/ml||1-2 Days||N/A|
|*Detection times should be considered estimates.|