Availability: In stock
only $110.75 when you buy 40 cases or more.
|Product sold in cases of 25 tests|
|$5.52 each||$4.92 each||$4.72 each||$4.56 each||$4.43 each|
|* Discounts may vary based on options above|
|HCDOAEW-2135||AMP / BAR / BUP / BZO / COC / mAMP / MDMA / MTD / MOP / OXY / PCP / TCA / THC|
|Stability Expiration Date||No|
|Application Type||urine cup|
|All Sales Final||No|
|Avg Competitor Price||$178.50|
PERFORMING THE TEST
Allow the test cup to come to room temperature [15-30℃ (59-86℉)] prior to test.
Step 1: Tear the foil bag open, remove test cup and disposable gloves provided for donor. Label the device with donor information. (Fig. 1)
Step 2: Open test cup lid. Urinary directly into the test cup. Be sure to fill up the test cup with the urine specimen between minimum 30ml to maximum 110ml (marked on the cup). (Fig. 2)
Step 3: After urine specimen has been collected, close the lid securely and return cup tp collection official. (Fig. 3)
Step 4: Collection official use glove provided. Peel off label to reveal test result. Read test result at 5 minutes. DO NOT INTERPRET RESULT AFTER 10 MINUTES. (Fig.4&5)
INTERPRETATION OF RESULTS
NEGATIVE: Two colored bands appear on the membrane. One band appears in the control region (C) and another band appears in the test region (T). Certain lines may appear lighter or thinner than other lines. A line is to be considered a line whether it is faint, light or dark.
*NOTE: The shade of red in the test line region (Drug/T) will vary, but it should be considered negative whenever there is even a faint pink line.
Preliminary Positive Result:
POSITIVE: Only one colored band appears, in the control region (C). No apparent colored band appears in the test region (T).
INVALID: Control band fails to appear. Results from any test which has not produced a control band at the specific read time must be discarded. Please review the procedure and repeat with a new test. If the problem persists, discontinue using the kit immediately and contact your local distributor.
STORAGE AND STABILITY
Store as packaged in the sealed pouch either at room temperature or refrigerated (2-30°C). The Test Device is stable through the expiration date printed on the scaled pouch. The Test Device must remain in the sealed pouch until use. Keep away from direct sunlight, moisture and heat. DO NOT FREEZE, Do not use beyond the expiration date.
QUESTIONS AND ANSWERS
Question 1: What does the Drug of Abuse Urine Test do?
Answer: These tests indicate if one or more prescription or illegal drugs are present in urine. The testing is done in two steps. First, you do a quick at-home test. Second, if the test suggests that drugs may be present, you send the sample to a laboratory for additional testing.
Question 2: What is “cut-off level”?
Answer: The cut-off level is the specified concentration of a drug in a urine sample. Above that concentration the test is called positive, and below that concentration it is called negative.
Question 3: What are drugs of abuse?
Answer: Drugs of abuse are illegal or prescription medicines (for example, Oxycodone or Valium) that are taken for a non-medical purpose, including taking the medication for longer than your doctor prescribed it for or for a purpose other than what the doctor prescribed it for.
Question 4: How accurate is the test?
Answer: The tests are sensitive to the presence of drugs in urine sample.These tests are not as accurate as lab tests. In some cases, certain foods and drugs may cause false positives as well as false negatives for those who use drug-testing kits.
Question 5: Does a preliminary positive screen test mean that you have found of abuse?
Answer: This means that the test has reacted with something in the sample and the sample must be sent to the lab for a more accurate test.
Question 6: What should I do, if the lab test confirms a positive result?
Answer: If you have received a confirmed positive result, please consult with our staff on a proper course of action. We will help you identify counselors who can help you. It is important that you remain calm and do not react in a negative way to the situation. If you do not believe the test result, please consult with your physician. They will have your background medical history and be able to provide you with detailed information on both the test and the meaning of the result.
|HCDOAEW-2135||AMP1000, BAR300, BUP10, BZO300, COC300, mAMP1000, MDMA500, MTD300, MOP300, OXY100, PCP25, TCA1000, THC50|
|Drug Name||ABBR||Calibrator||Standard Cutoff Levels Urine||Standard Cutoff Levels Saliva||Approx. Detection Time Urine||Approx. Detection Time Saliva|
500 & 1000 ng/ml
150 & 300 ng/ml
2-4 Days (Single)
300 & 500 ng/ml
Up to 80 Hours
200 & 300 ng/ml
K2 / Spice
2-5 Days (Single)
500 & 1000 ng/ml
*The above chart gives approximate detection periods for each substance by test type. The ranges depend on amount and frequency of use, metabolic rate, body mass, age, overall health, drug tolerance, urine pH, and other additional factors