Availability: In stock
|Product sold in cases (25 tests per case)|
|$6.69 each||$6.19 each||$5.95 each||$5.65 each||$5.35 each|
|* Discounts may vary based on options above|
|SSO-164||AMP / COC / mAMP / OPI / PCP / THC|
|Avg Competitor Price||$200.70|
DIRECTIONS FOR USE
Instruct the donor to not place anything in the mouth including food, drink, gum, and tobacco products for at least 10 minutes prior to collection.
Step 1: Remove the test device from the pouch and use it as soon as quickly as possible.
Step 2: Remove the collector from the sealed pouch. Take out of the collector cap; Insert the sponge into the mouth. Close mouth and move the sponge around for oral fluid collection. Soak sponge in oral fluid and swab the inside of the mouth and tongue. Collect oral fluid for 3 minutes until sponge is soft and fully saturated, No hard spots should be felt on the sponge when saturated.
Step 3: Place the test device on a clean and level surface. Remove the collection sponge from the mouth and insert it sponge first into the screening device, screw the collector cap locks in place in the bottom of the device. (See illustration 2）
Step 4: Test device upright on flat surface and keep upright while test is running. Waitfor the colored signal to appear in test results area. Read the results at 10 minutes.
INTERPRETATION OF RESULTS
NEGATIVE: Two colored bands appear on the membrane. One band appears in the control region (C) and another band appears in the test region (T). Certain lines may appear lighter or thinner than other lines. A line is to be considered a line whether it is faint, light or dark.
*NOTE: The shade of red in the test line region (Drug/T) will vary, but it should be considered negative whenever there is even a faint pink line.
Preliminary Positive Result:
POSITIVE: Only one colored band appears, in the control region (C). No apparent colored band appears in the test region (T).
INVALID: Control band fails to appear. Results from any test which has not produced a control band at the specific read time must be discarded. Please review the procedure and repeat with a new test. If the problem persists, discontinue using the kit immediately and contact your local distributor.
STORAGE AND STABILITY
Store as packaged in the sealed pouch at room temperature. The test device is stable through the expiration date printed on the sealed pouch.
The test device must remain in the sealed pouch until use. DO NOT FREEZE. Do not use beyond the expiration date.
|SSO-164||AMP50, COC20, mAMP50, OPI40, PCP10, THC12|
|Drug Name||ABB||Cutoff Level||Approx Detection|
time in Urine
time in Saliva
|Amphetamine||AMP||500 ng/ml||2-4 Days||1-3 Days|
|1000 ng/ml||2-4 Days||N/A|
|Barbiturates||BAR||300 ng/ml||3-6 Days||1 Day|
|Benzodiazepines||BZO||300 ng/ml||3-7 Days||1-2 Days|
|Buprenorphine||BUP||10 ng/ml||2-3 Days||1 Day|
|Cocaine||COC||150 ng/ml||2-4 Days||N/A|
|300 ng/ml||2-4 Days||1-3 Days|
|Ecstasy||MDMA||500 ng/ml||1-3 Days||1-2 Days|
|Fentanyl||FEN||300 ng/ml||3 Days||N/A|
|K2 / Spice||K2/Spice||50 ng/ml||1-3 Days||N/A|
|Marijuana||THC||50 ng/ml||5-30 Days||1 Day|
|20 ng/ML||Up To 30 days||N/A|
|Methadone||MTD||50 ng/ml||3-5 Days||1 Day|
|Methamphetamine||MAMP||1000 ng/ml||3-5 Days||1-3 Days|
|500 ng/ml||3-5 Days||N/A|
|Morphine||MOP||300 ng/ml||2-4 Days||1-3 Days|
|Opiates||OPI||2000 ng/ml||2-4 Days||1-3 Days|
|Oxycodone||OXY||100 ng/ml||2-4 Days||1 Day|
|Phencyclidine||PCP||25 ng/ml||5-10 Days||1-3 Days|
|Tricyclic Antidepressants||TCA||1000 ng/ml||3-6 Days||N/A|
|Propoxyphene||PPX||300 ng/ml||1-2 Days||N/A|
|*Detection times should be considered estimates.|